2019-09-24 10:02:48 Post time: 2019-09-24 10:02:48
思瑞三坐标 Author: Coordinate SEREIN
思瑞三坐标 Source: Siri Three Coordinates
1690 Attention: 1690
The appearance quality and internal structure of the car have become the focus of attention of various manufacturers. The precision of the check fixture in the quality control of the car is related to the matching and overlap of the product parts during the commissioning stage. As a "supervisor", CMM is on duty on duty, and strictly escorts the inspection tools from assembly to shipment. Once tested, unqualified products are immediately "hit the road" and re-adjusted and tested.
Gage debugging and testing, carefully coordinated three coordinates at each step
Taking a conventional automobile check fixture as an example, the testing process is: assembly , initial adjustment , fine adjustment , implantation of positioning pins , re-testing , exporting a test report and saving the measurement program . From the beginning of assembly to the completion of the commissioning of the three coordinates, each link quantifies the data and ships with the report.
安装 1 installation
Check all parts before installation, including appearance, burrs, chamfers and key parts size inspection.
调试 2 commissioning
：首先使用初调孔调好基准孔位置，然后手动取点与数模对比，借助三坐标读数观察数据变化，进行位置调整，初调一般调到公差附近越小越好 。 Initial adjustment method : First use the initial adjustment hole to adjust the reference hole position, and then manually take the point and compare with the digital model, observe the data changes with the three coordinate readings, and adjust the position. Generally, the initial adjustment is as small as possible near the tolerance.
：在3D数模上取点，执行程序自动测量目标点，得出偏差值进行调整，重复调整到公差范围内。 Fine tuning method : take points on the 3D digital model, execute the program to automatically measure the target point, get the deviation value to adjust, and repeat the adjustment to within the tolerance range.
Debugging "knock blackboard" knowledge points
：常见的调试顺序是先将检具的基准孔调试到公差范围内；目前使用的基准有：圆孔，圆柱，球，基准块。 Debugging sequence : The common debugging sequence is to first debug the reference hole of the fixture to within tolerance; the currently used standards are: round hole, cylinder, ball, reference block.
：基准孔要求最高（± 0.02mm ），定位孔次之（± 0.05mm ），齐平面与间隙面要求最低（± 0.1mm ）。 Accuracy requirements : The reference hole requires the highest (± 0.02mm ), the positioning hole takes the second place (± 0.05mm ), and the flush and clearance surfaces require the lowest (± 0.1mm ).
：大多数检具是使用4圆迭代先求出4个基准孔的实际坐标值，然后带入实测值再次迭代作为基准坐标对其他位置进行调试。 Debugging method : Most gages use 4 circle iterations to first obtain the actual coordinate values of the 4 reference holes, and then bring in the actual measured values and iterate again as the reference coordinates to debug other positions.
：定位孔的调试，敲击的力度，螺丝锁紧的手法 Difficulties in debugging: debugging of positioning holes, percussion strength, and screw locking methods
植入定位销 3 implant positioning pins
After qualified debugging in all positions, it is necessary to perform pin positioning on the connection part of the fixture. A common problem with plant pins is the misalignment of the pin hole . In this case, the pin hole is usually enlarged by one level. Generally, the pin hole is slightly smaller than the positioning pin. Use the interference fit to ensure the positioning of the inspection fixture. It is strictly forbidden to use glue to fix it. After the positioning is completed, the nameplate is attached and the surface is cleaned.
小时，然后编写自动测量程序，执行完成，完全合格后将检具放到周转仓库准备出货。 After the fixture has been planted and pinned, the nameplate is affixed, and clean-up is required to perform a full-scale three -dimensional re-test . At this time, the finished fixture must be thermostated for about 8 hours, and then an automatic measurement program must be written. After the execution is completed, the fixture Put it into the revolving warehouse for shipment.
Save programs and measurement reports
After the measurement is completed, the measurement program needs to be named and archived according to the gage number, a gage report is issued according to the customer's requirements, and the gage report is attached to the gage and shipped together.
Generally, the coordinates of the reference hole of the test fixture are placed on the first page, and then the pages are displayed in order according to the order of positioning holes, flush planes, and clearance surfaces.
For large three-dimensional coordinate measuring machines, install two display screens on the host computer, and then copy the screens. The two display screens have fewer dead angles, and you can observe the debugging effect from multiple directions.
Utilize the cycle command of the program and the pause button on the handle to realize the automatic operation of the machine controlled by the handle. After the debugging is completed, the computer is switched to the next position for debugging.
Gage detection "Wulin Cheats"
It is particularly troublesome for one person to debug large-scale fixtures. The following suggestions can effectively reduce the workload and improve the efficiency of debugging.
Gage detection is step by step under the real-time supervision of Sirui measuring three-coordinates, and it is stable and steady, helping customers achieve quality and efficiency.