Various indicator lights and meters are integrated on the car dashboard, which is an important device to reflect the working conditions of various systems of the vehicle. A lot of complicated inspection work is needed in the production process. Manual inspection methods have long been unable to keep up with automated production lines. The introduction of image measuring instrument detection is unstoppable.
Due to the numerous testing items and complex testing requirements of automotive dashboards, this requires fast testing speeds, high accuracy, and strong stability, which is really a great challenge for image measuring instrument testing. Data is the criterion for the feasibility of a solution. 全自动影像测量仪作为调研机型，加上特殊配置：探针，软件： SVM3.1 。 We use SVM 4030 DCC Standard automatic image measuring instrument as the research model, plus special configuration: probe, software: SVM3.1 . Perform precision inspection on the plastic parts of the instrument panel.
I. Customer's measurement requirements
Mainly measure the outer length, outer width and hole position.
Second, the clamping method
The product is relatively easy to fix. This time we use plasticine to fix it.
It is recommended to use imaging professional flexible fixture to fix the product.
Fourth, the detection process
Establish a coordinate system:
Coordinate system positioning using circles
The pure white or pure black of the product will reflect light. After turning on the gray mode, the rational effect of the measurement will be achieved.
V. Screenshots of the report:
Product dimensions are accurate and within tolerances.
Customer products can meet the measurement requirements using image measuring instruments. Since the height and side dimensions need to be measured, probes and prisms need to be added. Some of our customers' products are high and require objective lens measurement. 系列大行程影像测量机，测量效果效果更佳。 Since the products are automotive products, some products are too large, you can also use Siri SVM DCCCLASSIC series of large-stroke image measuring machine, the measurement effect is better. The choice of the image measuring instrument is determined by the efficiency requirements and product size.